2011 Annual Report
1. Distance-measuring unit - This unit measures the vertical distance between soil surface and the frame, on which the laser sensor is mounted. An “Acquity” sensor, which uses a fixed infrared laser and a rotating mirror to scan the soil surface along a straight line, will be used for distance measurement. In addition to distance measurement, this sensor also provides information on reflected light intensity. Thus, it is possible to distinguish between the top of canopies and actual soil surface using signal processing if soil surface is not completely covered by the canopy.
2. Frame and rail unit - Since the laser sensor only measures elevations along a straight line, a rail is needed to move the sensor in the direction perpendicular to the scan line so that elevations within a rectangular area can be measured. The rail will be supported by a frame. The sensor will travel along the rail, driven by a linear actuator; and the position of the sensor, monitored using an optical encoder, will be used as a feedback control signal to accurately control the sensor position. Several other devices, including a gyroscope to measure the angular position of the frame and an RTK GPS unit, will also be mounted on the frame.
3. Frame angular-position measuring unit - Because the laser-distance sensor will be mounted on the frame, angular displacements of the frame become critical to the accuracy of elevation measurement. Angular displacements of the frame - pitch, roll, and yaw - will be measured using a rate-integrating gyroscope. X, Y, Z coordinates of the laser scan lines on the soil surface will then be corrected using these measured angles.
4. Geo-reference unit -A Real-time Kinematic (RTK) GPS will be used to help register the measured surface points into a geographic coordinate system (UTM, Lat-lon, or a local coordinate system). An RTK GPS unit is needed because this is the only GPS device that provides a sub-centimeter accuracy in longitude and latitude.
5. Data-acquisition and control unit –All control and data signals from the laser, gyroscope, optical encoders and RTK GPS unit will be processed using a laptop computer.
Specific laboratory experiments were conducted with the laser distance meter to determine the accuracy and feasibility of measuring flat residue (wheat straw, corn stover, milo stalks, soybean residue, etc.) on various soil surfaces under different moisture conditions. Data were collected for comparison purposes using the laser system, a high resolution digital camera and a Sony camcorder with the infrared “nightshot” feature enabled. The laser system software was rewritten and modified to make it more robust and easier for a user to learn and correctly operate. In addition, the system wiring was replaced and more carefully routed along the rail to make it a more robust system in the field. The original rail motor was also replaced with a stepper motor and corresponding controller for more precision and control when moving the laser along the rail.
ADODR monitoring activities include phone calls, meetings, conference calls, and on-site visits.